Damaged building of the main building: a crack in the outer brick wall; the roof of the tent is cut by debris; window frames are broken, glazing of window frames is destroyed; door panels and boxes throughout the building have significant damage (broken, pierced by fragments, broken glass); minor damage to the interior decoration of the premises.
As a result of the blast wave, the equipment was damaged:
refrigerated centrifuges for separating donor blood (2 pcs.) – cracked housing, damaged rotor shaft; automatic device for blood sampling (scale-mixer – 1 pc.) – the case is damaged; laboratory binocular microscope (2 pcs.) – damaged optics;
hematological analyzer automatic Mindray 2800 (1 pc.) – does not work due to an explosive wave;
low-temperature refrigerators (3 pcs.) – damaged motors-compressors.